This post lists the background material of the hands-on tutorial about high-resolution EPI on SIEMENS scanners.
The MP2RAGE sequence is very popular for 7T anatomical imaging and is very commonly used to acquire 0.7-1 mm resolution whole brain anatomical reference data. Aside of this common application, it can also be very helpful for layer-fMRI studies to obtain even higher resolution T1 maps in the range of 0.5mm iso. However, when optimizing MP2RAGE sequence parameters for layer-fMRI studies, there are a few things that might be helpful to keep in mind.
In this post, I would like to discuss the challenges of using the popular MP2RAGE sequence in layer-fMRI studies. Specifically I will discuss challenges/features regarding:
- Background noise.
- Foldover artifact at very high resolutions.
- Usage of double-partial Fourier imaging.
- Inflow into arterial vessels.
In this blog post I want to go through the analysis pipeline of layer-dependent VASO.
I will go through the all the analysis steps that need to be done to go from raw data from the scanner to final layer profiles. The entire thing will take about 30 min (10 min analysis and 20 min explaining and browsing through data).
In this blog post, I want to share my thoughts on the number of layers that should be extracted from any given dataset. I will try to give an overview of how many layers are usually extracted in the field, I’ll describe my personal choices of layer numbers, and I will try to discuss the challenges of layer signal extraction along the way.
In this blog post Sri Kashyap and I describe how to deal with the registration of high-resolution datasets across days, across different resolutions, and across different sequences.
I am particularly fond of the following two tools: Firstly, ITK-SNAP for visually-guided manual alignment and secondly, using ANTs programs: antsRegistration and antsApplyTransforms.
Maximum intensity projection and minimum intensity projection can be insightful for mapping of vessels in 3D-slabs. In this post, I describe the application of intensity projections with LAYNII.
CBV-fMRI with VASO is highly dependent on a good inversion contrast. It gives it its CBV sensitivity and is also responsible for most of the VASO specific pitfalls (e.g. inflow, CSF etc. ). And thus, it should be optimized as much as possible.
In this blog post, I want to describe the most important features of a reliable inversion pulse for the application of VASO at 7T with a head transmit coil.